Experimental Physics and
We are using RS232. Using the XFlash unit as configured through spec does not show the same delay as the IOC. Disabling the Temperature monitoring PV (:ReadTemperature) which is on a 1s scan-rate, seems to alleviate the problem a bit.
On 2/6/13 11:00 AM, Phillip Sorensen wrote:
> We are tring to setup the Rontec XFlash MCA with epics using the R7.2
> version of mca.
> The problem we are having is there seems to be a delay of some number of
> seconds after sending a command until we see an update in the MEDM
> screen. For example clicking the start and stop buttons for the
> acquisition show this. I we look at the XFlash unit it seems to update
> the front panel immediately, but the MEDM screen shows a delay. The
> spectrum update (which is set to 1 second) also seems to take 5 to 10
> seconds per update.
> We are not getting any timeout errors. Any suggestions?
Over what physical link are you communicating with the XFlash? If it's
RS-232, it could just take that long to transmit the data over the slow
|ANJ, 20 Apr 2015||
· EPICS V4 · IRMIS · Talk · Bugs · Documents · Links · Licensing ·